MatrixNeuro Ultra Low Profile (ULP)
The MatrixNeuro system was developed in 2008 to offer a complete system for use in craniotomies, cranial trauma repair, and reconstruction of the craniofacial skeleton. Although it provided the lowest profile of plate and screw of its kind at that time, patients continued to require an even lower profile. Many patients find that plates can still be felt or seen as lumps under the skin. This is increasingly significant due to the growing number of procedures that include the frontal area, which, according to Prof Christian Matula, Chair of the Neuro Trauma Reconstruction Expert Group (NTREG) "is an area that the patient looks at every day in the mirror and any abnormality can make them feel unhappy". In addition, the unconscious rubbing of plates can lead to other skin issues.
The new MatrixNeuro Ultra Low Profile (ULP) system (Fig 1) was developed in response to the need to decrease implant profile whilst also providing comparable construct stiffness and strength to the plates in the existing MatrixNeuro system.
The plate thickness is only 0.28 mm (MatrixNeuro standard plates are 0.40 mm). Also, the screw recess was improved (Fig 2) to achieve a better seating of the screw, which has led to a plate-screw profile of 0.44 mm, a reduction of another 10% compared with the existing system. And instead of having a rectangular edge, the plates are chamfered to achieve almost no palpability under the skin (Fig 3).
The new MatrixNeuro ULP system is fully compatible with the existing MatrixNeuro system so that no new or specific instruments are needed, and the same surgical techniques are used. Furthermore, all implants are fixated with the already existing self-drilling screws from the MatrixNeuro system, which have a unique thread design for rapid screw starting and significantly lower insertion torque. The screws start with bone contact almost instantaneously providing a fast closure of bone flaps and rapid fixation of cranial fractures.
A wide range of plates is available (Fig 4) based on the most frequently used MatrixNeuro standard plates, ie, Burr Hole Covers ranging from 1224 mm with one additional version for shunt or drainage, as well as straight plates with center space in 2-hole and 4-hole versions. The system is completed with an X Plate, a square and a rectangular Frame Plate, a Double Y Plate, and a Strut Plate.
Case: Apert Syndrome
An 8-year-old girl with Apert Syndrome (Fig 5) was examined after being treated at another institution. The cause for loss of cranial bone was unknown. An anterior and posterior cranial expansion was given to treat the increased intracranial pressure.
After that healed, a simultaneous PSI cranioplasty and Le fort II osteotomy with zygoma repositioning were performed. The PEEK implants were fixated using MatrixNeuro ULP implants (Fig 6). A midface distraction device was then mounted on the PSI implants and distracted the Lefort 2 segment to treat the patient's sleep apnea (Fig 7).
Case provided by Richard Hopper, Seattle, USA.
Optimizing Surgical Cranial Access
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
“approved by AO Technical Commission” and “approved by AO Foundation”
The brands and labels “approved by AO Technical Commission” and “approved by AO Foundation”, particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.
AO TC Innovations Magazine
Find all issues of the AO TC Innovations Magazine for download here.
Recognizing outstanding achievements in development and fostering excellence in surgical innovation.