LCP Periarticular Aiming Instruments
The LCP periarticular aiming instruments are indicated to facilitate percutaneous, submuscular insertion of the LCP condylar 4.5 and LCP proximal tibia 4.5.
The set consists of specific aiming arms and handles designed to percutaneously aim for screw holes of these plates as well as common instruments such as drill sleeves, drill bits, guide wires, and connection parts.
The aiming arms will allow for percutaneous aiming of the LCP proximal tibia plate 4.5 up to 14 holes and the LCP condylar plate 4.5 up to 18 holes straight or curved.
The new aiming instruments feature several changes compared to the previous generation with the ultimate goal to improve handling. The aiming arms have been changed from PEEK to carbon fibre. The connection points have been redesigned to ensure there is no play or loosening in the aiming arm connection with the handle and plate. Snap-in sleeves through the aiming arms eliminate the use of thumb screws. The instrument color coding and labeling has been made more userfriendly.
It is intended to use similar instruments for all future 4.5 periarticular plate aiming systems.
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
“approved by AO Technical Commission” and “approved by AO Foundation”
The brands and labels “approved by AO Technical Commission” and “approved by AO Foundation”, particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.