ArcoFix and Synex II: the AOSpine Anterior Fracture Solution
AOSpine type A fractures of the thoracolumbar spine account for 66% of all spine injuries. They often cause an initial or, if left untreated, a subsequent kyphotic deformity. Especially the type A2 and A3 fractures commonly require a fusion. Furthermore, these fractures are a symptom of osteoporosis. A very accurate reconstruction and a strong mechanical support is crucial for a good outcome.
With the ArcoFix/Synex II combination, the AOSpine TK developed straight forward instrumentation for the minimal invasive anterior stabilisation of thoracolumbar tumors and fractures.
The development of this combined system was based on the following requirements:
- Implant and instrument must allow a maximum intraoperative versatility with regard to the realignment of the spine (kyphosis correction).
- Ease of use with a short learning curve.
- A less invasive procedure.
- Compatibility with previously developed technology.
- Respecting the AO philosophy for fracture treatment.
ArcoFix basically consists of three parts. The telescopic part allows for distraction and compression and the two swivelheads that enable a kyphosis or lordosis correction. Thanks to these unique features, the system allows the surgeon to first insert, position and fix the implant (vertebral body screws) and then perform the kyphosis correction and height restoration before everything is finally tightened (golden screws) and secured.
After the fracture is reduced and stabilized, the Synex II expandable cage is introduced with the angled Synex II spreader tip.
Once Synex II is in place it is expanded until the implant and the vertebral endplates are in full contact. Synex II is a modular system. Eight different central bodies can be combined with 15 different types of endplates. This helps to reach an ideal cross-sectional load distribution and thus to provide a very good mechanical support of the previously achieved correction.
The central body of Synex II was slimmed down compared to the Synex I. This leaves more space to be filled with more bone thus promoting fusion.
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
“approved by AO Technical Commission” and “approved by AO Foundation”
The brands and labels “approved by AO Technical Commission” and “approved by AO Foundation”, particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.