Internal Midface Distractor System
The Internal Midface Distractor provides an internal distraction device to correct severe midface retrusion and to treat patients born with craniofacial anomalies that result in severe midface deficiencies, including Apert Syndrome, Crouzon Syndrome, and Pfeiffer Syndrome. These syndromes produce severe midface retrusion and severe intracranial pressure which can lead to upper airway obstruction, degenerative vision, ulcerations of the cornea, potential blindness, sleep apnea, and psychosocial disorders.
The Internal Midface Distractor System provides several advantages in treating midface deficiencies over conventional osteotomies and bone grafting. For example, a LeFort III osteotomy in combination with bone grafts and rigid internal fixation is limited to approximately 10 mm of advancement due to the magnitude of the resulting soft-tissue forces resisting the advancement. Patients with severe midface retrusion of up to 40 mm can be treated with the Internal Midface Dis-tractor System in a single procedure because substantial soft-tissue forces exerted on the midface can be overcome.
The Internal Midface Distractor System also provides the ability to minimize relapse associated with the advancement. Employing distraction to correct these deformities provides a slow, controlled means for overcoming the forces of soft-tissue resistance which leads to an expectation (supported by clinical studies ) of a lesser degree of relapse. The presence of a buried distractor during bone consolidation further provides a rigid construct to counteract the soft-tissue forces until the bone has fully consolidated.
The Internal Midface Distractor System consists of a telescoping barrel attached to an anterior and a posterior footplate. It is placed subcutaneously with the anterior footplate fastened to the lateral orbital rim, extending down to the maxilla and spanning the zygomaticomaxillary suture, and the posterior footplate fastened to the temporal region of the cranium. The barrel of the distractor body lies sagittally against the cranium underneath the temporalis muscle, and its posterior end extends percutaneously into the external environment for activation. In a typical case, two devices are attached (one on each side) for advancement of the midface.
The outer barrel of the distractor body is threaded to accept both a hex nut and the posterior footplate. The posterior footplate is a fifteen-hole plate that can be moved anywhere along the threaded portion of the distractor body in 0.5 mm increments. The hex nut fixes the position of the posterior footplate once it has been chosen. The anterior foot plate attaches to the distractor body by a tongue and groove connection. This allows the surgeon to select from amongst five different styles of anterior footplates to fit the patients anatomy. A small machine screw can be used to rigidly secure the anterior footplate to the distractor body, or, if the machine screw is not used, the distractor body can be easily separated from the anterior footplate during removal without performing a full coronal incision. This enables a quick removal procedure of the distractor while leaving the anterior footplate implanted.
The device is offered in one length for up to 40 mm of distraction. The activation end of the device extends into the external environment on smaller patients, or, in the case of larger patients, an optional extension can be attached by a machine screw to lengthen the device. The extensions are offered in three versions: a rigid extension (20 mm length), an extension incorporating a universal joint (20 mm length), and an extension incorporating a flexible cable (40 mm length). The activation end of the distractor and the extensions engage with the patient screwdriver for advancements in 0.5 mm increments.
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
“approved by AO Technical Commission” and “approved by AO Foundation”
The brands and labels “approved by AO Technical Commission” and “approved by AO Foundation”, particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.