External midface distractor system
The external midface distractor is a distraction osteogenesis device that attaches to the cranium and midface and is used to gradually lengthen the midface at the Le Fort I, II, and III levels (including monobloc). It addresses the clinical need for an external distraction device to treat patients with congenital craniofacial anomalies with severe midface retrusion, which can lead to airway obstruction and nutrition difficulties. It meets requests for an external device to achieve controlled advancement of the midface where internal devices are not desirable. Therefore, it is particularly well suited for younger patients who are not skeletally mature and have mixed dentition. It can be placed and removed more easily than its internal counterpart and allows for postoperative vector control of the bone segments.
Some of the most common syndromes that are treated with the external midface distractor are Pfeiffer syndrome, Apert syndrome, and Crouzon syndrome. It can also be used on patients with cleft lip and palate as a secondary procedure in their teenage years to address their anterior maxillary growth deficiency.
The external midface distractor consists of an external head frame, a vertical carbon fiber rod assembly, horizontal crosspieces containing distraction screws, and separate internal footplate assemblies that attach to the zygoma and maxilla. The external components are connected to the internal hardware with wires. Scalp fixation pins are used to attach the device to the lateral aspects of the cranium.
The device is assembled prior to placement on the patient. The head frame, vertical carbon fiber rod assembly, and horizontal crosspiece with distraction screws are joined together and secured by set screws. For Le Fort I and II advancement one horizontal crosspiece is joined to the vertical carbon fiber rod assembly primarily at the level of the maxilla. For Le Fort III and monobloc advancement two horizontal crosspieces are joined to the vertical rod assembly; one primarily at the level of the zygoma or infraorbital rim and one primarily at the level of the maxilla.
The head frame can be adjusted to fit a range of patient head sizes. The head frame expands laterally as a unit using a central adjustment screw located in its hub. Rear adjustment screws located in the mounting plates of the head frame adjust the anterior projection of the device. The vertical carbon fiber rod can be adjusted laterally for placement along the patients midline, independent of the head frame position on the skull. The vertical carbon fiber rod can also be rotated both anteriorly/ posteriorly and right/left to minimize obstruction of the patients path of vision and enable to the patients to have an unobstructed access to deliver food to the mouth.
Mounting pins in rounded, conical, and self-drilling versions are used to secure the device to the patient. Rounded pins do not pierce the soft tissue and are used for initial stabilization of the device on the cranium. Conical pins pierce the soft tissue to apply clamping pressure to the lateral skull bones for securing the device. Self-drilling pins are offered as an alternative to conical pins to engage and secure the headframe to the bone by threading into the bone in the same manner similar to standard cranial bone screws.
Active distraction is accomplished by the patient using a screwdriver that mates with a hex nut on each distraction screw assembly. The mid-face can be advanced in 0.5 mm increments by the patient. According to the surgeons instruction. Further refinements of the position of the midface are accomplished by the surgeon using the adjustment capabilities of the device. This is achieved by loosening the wires connecting the external device to the internal hardware, repositioning the horizontal cross pieces, and retightening the wires. Removal of the device is accomplished by cutting the wires between the internal and external hardware, removing the mounting pins from the cranium, and removing the internal hardware. Depending on the type of internal hardware used, a secondgeneral anesthesia may or may not be needed for device removal.
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
“approved by AO Technical Commission” and “approved by AO Foundation”
The brands and labels “approved by AO Technical Commission” and “approved by AO Foundation”, particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.
AO TC Innovations Magazine
Find all issues of the AO TC Innovations Magazine for download here.
Recognizing outstanding achievements in development and fostering excellence in surgical innovation.