LiLouPen (Light Loup Pen)
The change of screw mechanism from a hexagon socket to StarDrive combined with minimally invasive techniques of implant removal place new demands on the surgeons and their instruments.
In order to remove the screws it is important to identify the size and the type of screw mechanism without having to expose the implant by surgical dissection.
The size of the screw can generally be identified radiologically with reference to the anatomical region, but the screw mechanism generally cannot.
The LilouPen has been designed to meet this need. It is the realization of a straightforward endoscopic procedure for the purpose of identifying the screw mechanism, freeing the screw head from ingrowing soft tissues, and positioning the screwdriver.
The pen is made from a transparent material, either glass or synthetics. Glass has the advantage that an optical reference (hologram) can be recorded. A synthetic material permits simple surface treatment in the sense of a Fresnel lens for light input. In a bright operative environment the image needs to be illuminated and displayed with as little dissipation as possible. The light for the illumination comes from sources in the environment (theater lights) or from a small LED attached to the pen.
A cylindrical retractor is used to expose the field of view, whereby the soft tissues between the end of the pen and the implant can be pushed away radially. The retractor consists of an expandable, cylindrical sleeve. The LilouPen is especially useful when the implants are covered by a thick, soft-tissue layer.
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
“approved by AO Technical Commission” and “approved by AO Foundation”
The brands and labels “approved by AO Technical Commission” and “approved by AO Foundation”, particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.
AO TC Innovations Magazine
Find all issues of the AO TC Innovations Magazine for download here.
Recognizing outstanding achievements in development and fostering excellence in surgical innovation.